Trials of a new relative humidity sensor
Payne, Richard E.
MetadataShow full item record
A new relative humidity and air temperature sensor, the Sensirion Model SHTl, has been thoroughly tested by the Upper Ocean Processes (UOP) group at the Woods Hole Oceanographic Institution. One-minute averages from two of the sensors, as well as a Väisälä HMP4SA, were recorded for over a year. A third Sensirion sensor was kept in the laboratory and calibrated at monthly intervals with the other three sensors. The standard deviation of the difference in relative humidity between the Sensirion sensors and the Väisälä was about 2% RH. The difference in air temperature was about 0.2°C. Drift rates in relative humidity for the two Sensirion sensors were 2.7% RH/yr and -0.3% RH/yr, and in air temperature, O.1°C/yr and 0/3°C/yr. Because one of the two Sensirion sensors deployed outside had significant variations in its calibration, the UOP group will not adopt these sensors. However, their very small size, low-cost, and low-power requirements may make them desirable for other uses.
Showing items related by title, author, creator and subject.
Payne, Richard E. (Woods Hole Oceanographic Institution, 1994-09)An analysis has been made of the calibrations done during 1990 through 1994 on Rotronic MP- 100 relative humidity sensors (used in the Improved METeorological (IMET) system and sensors built at the Woods Hole Oceanographic ...
Improved meteorological measurements from buoys and ships (IMET) : preliminary comparison of humidity sensors Crescenti, Gennaro H.; Payne, Richard E.; Weller, Robert A. (Woods Hole Oceanographic Institution, 1990-05)Humidity sensors using various principles of operation are evaluated for the potential use at sea on buoys and ships. Thin film capacitive polymer sensors include the Vaisala Humicap HMP-14U (with WHOI electronics), Hy-Cal ...
Hosom, David S.; Winget, Clifford L.; Weisman, Sumner; Doucet, Donald P.; Price, James F. (Woods Hole Oceanographic Institution, 1988-12)A new, intelligent, chilled mirror humidity instrument has been designed for use on buoys and ships. The design goal is to make high quality dew point temperature measurements for a period of up to one year from an ...