An all-thermistor pyrgeometer
Payne, Richard E.
MetadataShow full item record
The design and testing of an all-thermistor (no thermopile) pyrgeometer (LWT) is descrbed. After calibration by comparison with a Kipp & Zonen CG4, 9.2 months of data show mean differences of order 1-2 W m-2 with standard deviations of order 7-8 W m-2. Approximately half of the mean difference and the standard deviation derived from 40 occasions when the LWT readings were anomalously high for periods of 2-l4 hours, principally at night. No reason has been found for the anomalous behavior. During the 9.2 months, of data, there were also 11 periods of a few hours each when the Eppley PIR indicated noticeably higher flux values than did the Kipp & Zonen CG4. The conclusion reached is that contemporary thermistors allow temperature measurements of suffcient accuracy, and the thermopile can be eliminated from pyrgeometers. The differences seen between the Kipp & Zonen and the Epply raise doubts about their absolute accuracies on time scales of hours, although their long-term averages are quite comparable.
Suggested CitationTechnical Report: Payne, Richard E., "An all-thermistor pyrgeometer", 2004-12, DOI:10.1575/1912/787, https://hdl.handle.net/1912/787
Showing items related by title, author, creator and subject.
Timmermans, Mary-Louise; Melling, Humfrey; Rainville, Luc (American Meteorological Society, 2007-04)A 50-day time series of high-resolution temperature in the deepest layers of the Canada Basin in the Arctic Ocean indicates that the deep Canada Basin is a dynamically active environment, not the quiet, stable basin often ...
Drifting vertical current meter, moored aanderaa thermistor chain, and XBT data : Jasin 1978 Atlantis-II cruise (102) Pennington, Nancy J.; Weller, Robert A. (Woods Hole Oceanographic Institution, 1981-10)The report presents summaries of three data sets taken at and in the vicinity of the oceanographic moorings deployed in the 1978 Joint Air-Sea Inte raction Project (JASIN). The data sets are: (1) the temperature, pressure ...
Temperature from a thermistor chain deployed along a 30m depth contour at Mission Beach, CA in June of 2016 Franks, Peter; Lucas, Andrew J (Biological and Chemical Oceanography Data Management Office (BCO-DMO). Contact: email@example.com, 2020-01-02)Temperature from a thermistor chain deployed along a 30m depth contour at Mission Beach, CA in June of 2016. For a complete list of measurements, refer to the full dataset description in the supplemental file ...