Integrated seawater sampler and data acquisition system prototype : final report
Berteaux, Henri O.
Eck, Calvert F.
Irish, James D.
Jenkins, William J.
Kery, Sean M.
Albro, Carl S.
MetadataShow full item record
This report documents the work performed by the Woods Hole Oceanographic Institution (WHOI) and the Battelle Memorial Institute from August 1988 to December 1992 in the NSF sponsored development of an Integrated Seawater Sampler and Data Acquisition Prototype. After a 6-month initial design study, a prototype underwater profiing unit was designed and constructed, containing the water acquisition subsystem, CTD and altimeter, control circuitry and batteries. A standard WHOI CTD was adapted for use in the underwater unit and was interfaced to the underwater controller which had a telemetry module connecting ít with a deck control unit. This enabled CTD data to be logged in normal fashion on shipboard while additional commands and diagnostics were sent over the telemetry link to command the underwater unit's water sampling process and receive diagnostic information on system performance. The water sampling subsystem consisted of 36 trays, each containing a plastic sample bag, the pump and control circuitry. The sample bags, initially sealed in a chemically clean environment, were opened by pumping the water out of the tray, thus forcing water into the bag by ambient pressure. The command system could select any bag, and control the water sampling procss from the surface with diagnostic information on system altitude, depth, orientation and cable tension displayed in real time for operator information. At sea tests confirmed the operation of the electrical and control system. Problems were encountered with the bags and seals which were partially solved by further post cruise efforts. However, the bag closing mechanism requires further development, and numerous small system improvements identified during the cruises need to be implemented to produce an operational water sampler. Finally, initial design tor a water sampler handling and storage unit and water extraction system were developed but not implemented. The detailed discussion of the prototype water sampler design, testing and evaluation, and new bag testing result are presented.
Suggested CitationTechnical Report: Berteaux, Henri O., Eck, Calvert F., Irish, James D., Jenkins, William J., Kery, Sean M., Albro, Carl S., McDowell, S., "Integrated seawater sampler and data acquisition system prototype : final report", 1993-04, DOI:10.1575/1912/624, https://hdl.handle.net/1912/624
Showing items related by title, author, creator and subject.
Culture-independent identification of bacteria present in the pressure-retaining seawater (PRS) sampler deployed during Leggo drop 1 from R/V Falkor cruise FK141215 in the Challenger Deep, Mariana Trench in December 2014 Bartlett, Douglas (Biological and Chemical Oceanography Data Management Office (BCO-DMO). Contact: email@example.com, 2020-01-21)Culture-independent identification of bacteria present in the pressure-retaining seawater (PRS) sampler deployed during Leggo drop 1. For a complete list of measurements, refer to the full dataset description in the ...
SyPRID sampler : a large-volume, high-resolution, autonomous, deep-ocean precision plankton sampling system Billings, Andrew F.; Kaiser, Carl; Young, Craig M.; Hiebert, Laurel S.; Cole, Eli; Wagner, Jamie K.S.; Van Dover, Cindy (Elsevier, 2016-05-19)The current standard for large-volume (thousands of cubic meters) zooplankton sampling in the deep sea is the MOCNESS, a system of multiple opening–closing nets, typically lowered to within 50 m of the seabed and towed ...
A self-powered pumping system for in situ extraction of particulate and dissolved materials from large volumes of seawater Winget, Clifford L.; Burke, John C.; Schneider, David L.; Mann, Don R. (Woods Hole Oceanographic Institution, 1982-02)A pumping system has been developed for the in situ extraction of particles and of dissolved constituents from large volumes of seawater. The assembly consists of a battery-powered submersible pump, filters, and ...