On the formulas for correcting reversing thermometers
Keyte, Freeman K.
MetadataShow full item record
The case often arises where a thermometer which has been inserted into a medium of temperature Tw is actually read in a place where the environment is at temperature t, ≠ Tw. Such a case is the soil thermometer , where the bulb is at Tw and the stem in the air at t; and such a case is the oceanographic reversing thermometer, brought up from a depth-of-reversal (Tw) to the ship laboratory (t). In each case the different cubical expansion coefficients of mercury and glass mean that the stem mercury capillary is taken from the true reading of Tw to a reading of T/ by the action of environmental change from Tw to T/.
Suggested CitationTechnical Report: Keyte, Freeman K., "On the formulas for correcting reversing thermometers", 1964, DOI:10.1575/1912/25229, https://hdl.handle.net/1912/25229
Showing items related by title, author, creator and subject.
Whitney, Geoffrey G. (Woods Hole Oceanographic Institution, 1979-08)The nomenclature, morphology and functioning of deep sea reversing thermometers are identified and explained. The various types of malfunctions, nonfunctions and other aberrations are defined and detailed instructions ...
Slavinsky, Mark; Bogolubov, Boris; Alelekov, Igor; Pigalov, Konstantin; Spiesberger, John L.; Boutin, Paul R. (Woods Hole Oceanographic Institution, 1993-02)A compact electromagnetic monopole source, requiring pressure equalization, was evaluated at the Naval Underwater Systems Center at Lake Seneca during July 1992 by scientists from the Institute of Applied Physics of the ...
Rodriguez, Luis G.; Cohen, Anne L.; Ramirez, Wilson; Oppo, Delia W.; Pourmand, Ali; Edwards, R. Lawrence; Alpert, Alice; Mollica, Nathaniel R. (American Geophysical Union, 2019-05-31)The Holocene is considered a period of relative climatic stability, but significant proxy data‐model discrepancies exist that preclude consensus regarding the postglacial global temperature trajectory. In particular, a ...