Software development for continuous-gas-flow AMS
von Reden, Karl F.
Roberts, Mark L.
Jenkins, William J.
Rosenheim, Brad E.
McNichol, Ann P.
Schneider, Robert J.
MetadataShow full item record
KeywordNumerical simulation; Data acquisition - hardware and software; Accelerator mass spectrometry
The National Ocean Sciences Accelerator Mass Spectrometry (NOSAMS) Facility at Woods Hole Oceanographic Institution is presently completing installation of a novel continuous-flow AMS system. A multi-year development of an AMS microwave gas ion source in collaboration with Atomic Energy Canada Limited (AECL), Chalk River, has preceded this final step of an implementation that is expected to add a new dimension to 14C AMS. National Instruments, NIM, and CAMAC modules have been programmed with LabVIEW on a Windows XP platform to form the basis for data acquisition. In this paper we discuss possible applications and include simulations of expected data acquisition scenarios like real-time AMS analysis of chromatograms. Particular attention is given to issues of synchronization between rapidly changing input amplitudes and signal processing cycles in hardware and software.
Author Posting. © Elsevier B.V., 2008. This is the author's version of the work. It is posted here by permission of Elsevier B.V. for personal use, not for redistribution. The definitive version was published in Nuclear Instruments and Methods in Physics Research Section B: Beam Interactions with Materials and Atoms 266 (2008): 2233-2237, doi:10.1016/j.nimb.2008.03.001.
Showing items related by title, author, creator and subject.
Bowlin, James B.; Spiesberger, John L.; Duda, Timothy F.; Freitag, Lee E. (Woods Hole Oceanographic Institution, 1992-10)A new computer program for accurate calculation of acoustic ray paths through a range-varing ocean sound channel has been written. It is based on creating a model of the speed of sound in the ocean, consistent with input ...
Montgomery, Ellyn T. (Woods Hole Oceanographic Institution, 1991-01)This report provides a User's Guide for operation of the High Resolution ProfIer (HRP) and documentation of the software structure and recent modifications. The HRP is an instrument that acquires and stores up to 16 types ...