Mechanical and operational details of a Neil Brown Instrument Systems internally recording conductivity, temperature, depth (CTD) profiler
Trask, Richard P.
MetadataShow full item record
The purpose of this report is to discuss the use of a Neil Brown Instrument Systems internal recording CTD. The components of the instrument are described along with the advantages and disadvantages of the internal recording system. Calibration of the pressure and temperature sensors in the laboratory and the method used for in situ calibration of the conductivity sensor is described. A step by step description of the use of the CTD/IR at sea is also included.
Suggested CitationTechnical Report: Trask, Richard P., "Mechanical and operational details of a Neil Brown Instrument Systems internally recording conductivity, temperature, depth (CTD) profiler", 1981-09, DOI:10.1575/1912/10275, https://hdl.handle.net/1912/10275
Showing items related by title, author, creator and subject.
Hess, Frederick R.; Orr, Marshall H. (Woods Hole Oceanographic Institution, 1979-10)The system described provides wide operational flexibility at any operating frequency from 5 kHz to over 800 kHz (except for a small band around 455 kHz) limited mainly by the availability of transducers. Variable pulse ...
Koehler, Richard L.; von Herzen, Richard P. (Woods Hole Oceanographic Institution, 1986-01)A miniature temperature recorder has been developed to be used with the hydraulic piston sediment corer <HPC) on the Deep Sea Drilling Project <DSDP). The instrumentation fits into pressure-sealed slots in the wall of the ...
Brown, Neil L.; Morrison, G. K. (Woods Hole Oceanographic Institution, 1978-02)A Conductivity, Temperature and Depth (CTD) profiler has been designed to make precise fine scale measurements of these physical parameters in the ocean. This CTD system consists of a shipboard Data Terminal deck unit ...