Improved meteorological measurements from buoys and ships (IMET) : preliminary comparison of humidity sensors
Crescenti, Gennaro H.
Payne, Richard E.
Weller, Robert A.
MetadataShow full item record
Humidity sensors using various principles of operation are evaluated for the potential use at sea on buoys and ships. Thin film capacitive polymer sensors include the Vaisala Humicap HMP-14U (with WHOI electronics), Hy-Cal Engineerig Ultra-H (also with WHOI electronics), the new Vaisala HMP-35A, and the Rotronic MP-lOOF. Impedance sensors include the Thunder Scientific PC-2101, Phys-Chem PCRC-ll, and the General Eastern 850. The Hygrometrix 8503A is the only organically based cellulose crystallite sensor evaluated. Chilled mirror dew sensors include the EG&G 200M Dewtrak, which was used as a comparative standard, the General Eastern Dew-lO and the WHOI D10IQ Intelligent Dew Point Sensor. The IR-200 infrared optical hygrometer from Ophir is also included in this study. The performance of the EG&G 200M Dewtrak was quite disappointing. Errors of up to 2.5°C in air temperature were observed due to inadequate shielding from solar radiation.
Showing items related by title, author, creator and subject.
Payne, Richard E. (Woods Hole Oceanographic Institution, 1994-09)An analysis has been made of the calibrations done during 1990 through 1994 on Rotronic MP- 100 relative humidity sensors (used in the Improved METeorological (IMET) system and sensors built at the Woods Hole Oceanographic ...
Payne, Richard E. (Woods Hole Oceanographic Institution, 2004-12)A new relative humidity and air temperature sensor, the Sensirion Model SHTl, has been thoroughly tested by the Upper Ocean Processes (UOP) group at the Woods Hole Oceanographic Institution. One-minute averages from two ...
Hosom, David S.; Winget, Clifford L.; Weisman, Sumner; Doucet, Donald P.; Price, James F. (Woods Hole Oceanographic Institution, 1988-12)A new, intelligent, chilled mirror humidity instrument has been designed for use on buoys and ships. The design goal is to make high quality dew point temperature measurements for a period of up to one year from an ...