Intelligent chilled mirror humidity sensor
Hosom, David S.
Winget, Clifford L.
Doucet, Donald P.
Price, James F.
MetadataShow full item record
A new, intelligent, chilled mirror humidity instrument has been designed for use on buoys and ships. The design goal is to make high quality dew point temperature measurements for a period of up to one year from an unattended platform, while consuming as little power as possible. Nominal system accuracy is 0.3°C, and a measure of data quality is provided to indicate possible drift in calibration. Energy consumption is typically 800 Joules per measurement; standby power consumption is 0.05 watts. Control of the instrument is managed by an onboard central processing unit which is programmable in BASIC, and communication to an external data logger is provided through an RS232 compatible interface. This report describes the preliminary sensor tests that led to this new design and provides the complete technical description required for fabrication.
Showing items related by title, author, creator and subject.
Krishfield, Richard A.; Toole, John M.; Proshutinsky, Andrey; Timmermans, Mary-Louise (American Meteorological Society, 2008-11)An automated, easily deployed Ice-Tethered Profiler (ITP) instrument system, developed for deployment on perennial sea ice in the polar oceans to measure changes in upper ocean water properties in all seasons, is described, ...
Sannino, Gianmaria; Pratt, Lawrence J.; Carillo, Adriana (American Meteorological Society, 2009-11)The hydraulic state of the exchange circulation through the Strait of Gibraltar is defined using a recently developed critical condition that accounts for cross-channel variations in layer thickness and velocity, applied ...
Kida, Shinichiro; Yang, Jiayan; Price, James F. (American Meteorological Society, 2009-02)Marginal sea overflows and the overlying upper ocean are coupled in the vertical by two distinct mechanisms—by an interfacial mass flux from the upper ocean to the overflow layer that accompanies entrainment and by a ...