An optical telemetry device to identify which dolphin produces a sound
Tyack, Peter L.
MetadataShow full item record
A small telemetry device, called a "vocalight," was designed for attachment to a dolphin's head using a suction cup. The vocalight lights up a variable number of light-emitting diodes depending upon the loudness of sounds received at a hydrophone within the suction cup. If vocalights matched for sensitivity are put on each dolphin within a captive group, observers can identify which dolphin produces a vocalization. Use of vocalights indicates that source levels of whistles from captive bottlenosed dolphins, Tursiops truncatus, range from approximately 125 to over 140 dB re: 1 µPa at 1 m.
Author Posting. © Acoustical Society of America, 1985. This article is posted here by permission of Acoustical Society of America for personal use, not for redistribution. The definitive version was published in Journal of the Acoustical Society of America 78 (1985): 1892-1895, doi:10.1121/1.392777.
Showing items related by title, author, creator and subject.
Acousto-optic effect compensation for optical determination of the normal velocity distribution associated with acoustic transducer radiation Foote, Kenneth G.; Theobald, Peter D. (Acoustical Society of America, 2015-09-28)The acousto-optic effect, in which an acoustic wave causes variations in the optical index of refraction, imposes a fundamental limitation on the determination of the normal velocity, or normal displacement, distribution ...
Siegel, David A.; Peterson, P.; McGillicuddy, Dennis J.; Maritorena, S.; Nelson, N. B. (American Geophysical Union, 2011-07-14)We investigate the bio-optical footprints made by mesoscale eddies in the Sargasso Sea and the processes that create them through an eddy-centric approach. Many (>10,000) eddies are identified and followed in time using ...
Bowen, Martin F. (Woods Hole Oceanographic Institution, 2001-06)Neutral tethers for unmanned underwater vehicles may contain a bundle of single-mode, optical fibers that are protected inside flexible stainless steel tubing. To date the author knows of no commercially available systems ...