The high speed buffer board : a SAIL EIA-485 communications accelerator card for the vector measuring current meter
MetadataShow full item record
A High Speed Buffer Board (HSBB) has been developed for the Vector Measuring Current Meter (VMCM) to implement the transmission of data at 9600 baud over an EIA-485 link. The HSBB significantly extends the VMCM communication functionality, which was previously limited to 300 baud transmission via 20 mA current loop or FSK telemetry. The increased speed allows rapid sampling of a large number of current meters on a common cable and the EIA-485 circuitry, which was designed for low power operation, provides a useful multipoint communication method for data transmission over long cable lengths. SAIL protocol (IEEE 997) was utilized to coordinate data transfer by the instruments on a common link. An MC68HC11 microcontroller resides in the VMCM, buffering data it receives at 300 baud from the VMCM UART. In response to a jumper selectable SAIL address, the MC68HC11 offloads the data at 9600 baud via EIA-485 to the SAIL controller. Synchronous data collection from many instruments is ensured by the SAIL synoptic set command and an embedded resynchronization/reset command. The low power consumption allows deployments of six months or more with a standard VMCM battery stack.
Showing items related by title, author, creator and subject.
Moored current meter data from the Atlantic north equatorial counter current near 6°N 28°W (February-September, 1983) : vol. XXXIV Levy, Ellen; Richardson, Philip L. (Woods Hole Oceanographic Institution, 1984-05)This report presents current and wind data from the first of three surface mooring deployments in the Atlantic North Equatorial Countercurrent near 6°N and 28°W. A Vector Averaging Wind Recorder (VAWR) measured wind ...
Panofsky, H. A. (Woods Hole Oceanographic Institution, 1951-09)The members of the Marine Meteorology Project of the Woods Hole Oceanographic Institution have designed equipment for the measurement or the vertical heat flux. This consists of a small vertical vane, a heated thermistor ...
Aubrey, David G.; Spencer, Wayne D.; Trowbridge, John H. (Woods Hole Oceanographic Institution, 1984-05)The dynamic response of electromagnetic current meters (manufactured by Marsh-McBirney, Inc.) has been clarified through a comprehensive laboratory measurement program combined with a thorough literature review. ...